The tip of the probe can then be changed to a locked 90˚ position to test the strength of a soft tissue repair or to probe defects.
|Working length||115 mm|
Cleaning and Sterilization
Remove gross soiling, blood and organic matter after surgery. Clean the instruments with sterile water (not saline).
Manual cleaning in enzymatic soak solution.
Ultrasonic cleaners – Fine cleaning (hollow and tubular instrument).
Wiped and dried with clean dry sponge before storage.
Care of the personnel handling instruments – aerosolization, splashing of infectious material, injury from sharp objects.
Instruments should be properly arranged in trays.
Sharp instrument tips should be covered with caps.
Instruments with moving parts should be lubricated after cleaning with water soluble lubricant.
Ethylene Oxide gas – good disinfectant
Steam autoclave – Probes, punches, graspers (mechanical instruments)
Arthroscope, light cable – Cidex, Steris (paracetic acid solution) – 30 mins
Use and Maintenance
|Use||Used for lifting and retracting tissues. The Probe enables the surgeon to easily insert an instrument through a small stab incision in soft tissue with the tip in a straight position.|